Distraction Osteogenesis Treatment & Management
- Author: Pravin K Patel, MD; Chief Editor: Jorge I de la Torre, MD, FACS more...
Regardless of which facial skeletal element is undergoing distraction, the treatment can be divided broadly into the following phases:
This phase involves radiographic studies to determine the feasibility of placement of the distraction device, whether an internal or external device is more appropriate, and the vector (direction, amplitude) of the distraction. Anticipated trajectory depends directly on the distraction vector, which can be vertical, horizontal, or oblique. Multiplanar devices even allow manipulation of the vector during the distraction phase. When possible, the use of printed, patient-specific 3-D anatomic models helps to visualize the placement of the distraction device and the osteotomy and to simulate the distraction process. See the image below.
Involvement of the orthodontist is essential during the presurgical phase. Presurgical orthodontic preparation assists the skeletal distraction by providing an occlusal guide.
Osteotomies used with distraction are well described with the conventional reconstructive approaches and need only be modified to accommodate the specific device. While the exact details may vary with the procedure, the following are guidelines:
Adequate mandibular bone stock must be available for the osteotomy and placement of the device.
Numerous factors should be considered when deciding between an internal versus external device. External devices allow for more predictable, multidirectional control of the distraction, which cannot be achieved with the currently available internal devices. However, external devices require multiple skin incisions that may lead to significant facial scarring. For many children and their families, the application of sequential distraction-vectors with a series of internal devices is preferable to the risk of permanent external scars.
The approach, either intraoral or extraoral, depends upon the degree of bony and soft-tissue exposure required for placement of the device and the allowable maxillary-mandibular opening.
The placement and direction of the device dictates the distraction vector. The osteotomy line does not necessarily need to be perpendicular to the distraction vector but should be placed to avoid injury to the inferior alveolar nerve and the developing dentition. In addition, avoidance of such injury can be facilitated by an incomplete osteotomy with subsequent separation occurring during the distraction phase. See the image below.
Temporarily fixing the distractor into position before making the osteotomy can simplify distractor placement. Positioning the device after the osteotomy can be difficult because of the mobility of the proximal segment.
Employ standard principles of a sagittal split osteotomy when lengthening the mandibular body. Preserve the nerve by using a reciprocating saw for the buccal corticotomy and “green-stick” fracturing the lingual cortex with an osteotome. Complete mobilization is not always necessary, since the distraction device completes the osteotomy. Warn the patient and family of the discomfort the patient will feel until the fracture is completed.
Prior to closure, test the device and clearly mark for the family the direction (clockwise or counterclockwise) of the driver used to turn the device.
Midfacial and frontofacial distraction
The use of external devices (head frame and/or helmet) typically requires the presurgical placement of a palatal appliance to guide the distraction vector in multiple planes relative to the skull base.
The midface must be completely mobilized as with conventional approaches with conventional, well-defined osteotomies. Avoid dental root disruption during the stages of primary or mixed dentition by modifying the typical Le Fort I osteotomy. Place the horizontal cut more cephalad, near the level of the inferior orbital foramen.
Midfacial advancements at the Le Fort I level with currently available internal devices are limited because of the difficulty of appropriately orienting the devices in the limited space. The fixation of the device may also injure the developing dentition. External multidirectional devices are preferred, as they allow more control over the vector of the distraction process.
Midfacial advancement at the Le Fort III level and frontofacial advancements can be approached with either internal or external devices, depending on the circumstances. Place the internal devices at the level of the body and arch of the zygoma. External devices require a palatal appliance and are supported by additional traction wires at the zygoma, nasal root, and supraorbital regions.
Before proceeding with distraction, there is a variable period (latency period) to allow for initial bone formation to occur. The period is typically 3-5 days, although in neonates and infants, the latency period may be omitted or last only 24 hours. With skeletal maturity, in contrast, the latency period is typically 5-7 days.
The process of distraction is activated when bone segments are gradually pulled apart using either an internal or external device. Three variables must be set: the rate of distraction, the rhythm or frequency of distraction, and the total time of distraction. The rate of distraction is typically 1.0 mm/d. Some advocate up to 2.0 mm/d in younger children to avoid early consolidation and a slower rate of 0.25-0.5 mm/d in older patients. This can be accomplished either once a day or divided throughout the day, determining the rhythm or frequency of distraction. While the distraction rate is 1.0 mm/d, ideally maintain the tissues under constant tension by dividing the total daily rate of distraction into smaller increments throughout the day to favor histogenesis.
The total time of the distraction phase is customized to the severity of the deformity and the patient’s demographics. There can be a discrepancy between the anticipated bone length and the total time of distraction. External devices that use pins to transmit the forces frequently bend, and the distance at the site of the distracting mechanism rarely equals the distance of the gap at the osteotomy sites. In hemifacial microsomia, for example, the position of the menton, distance from the lateral canthus to the commissure, and the mandibular cant should serve as clinical guidelines.
Once the desired correction is achieved with the distraction phase, allow mineralization of the immature bone to occur. Lock the distracting appliance into place to maintain stability until the newly formed bone has sufficient strength. The length of this phase varies depending on the circumstances. In general, 6-8 weeks is considered adequate. A guideline used by some centers is 2 days of consolidation to every day of distraction.
Remove the device and maintain stability, typically with the assistance of orthodontic appliances. In children with hemifacial microsomia, this step may require occlusal splints to guide the maxilla into position when the leveling of the mandibular cant creates a posterior open bite. In children with midfacial deformity, retention may require a face mask with elastic traction for a period of time.
Complications specific to the distraction process include the following :
Device mechanism failure
Injury to the developing tooth follicles (eg, maxillary and mandibular osteotomies)
Injury to various branches of the facial or trigeminal nerves (eg, the inferior alveolar nerve with mandibular distraction)
Pin site infection with external devices or semiburied devices
Nonunion and premature fusion
Complications specific to the osteotomy
Psychosocial issues related to the recovery (length of treatment time and patient's physical appearance with distraction device)
Outcome and Prognosis
With increasing clinical experience, the long-term outcome and the specific role of distraction osteogenesis are better defined today. Clearly, distraction can generate bone with the capacity for remodeling and adaptation to functional loads. However, distraction osteogenesis is likely incapable of restoring the normal development of a once dysplastic pattern of craniofacial growth. Distraction techniques allow the surgeon to intervene earlier in childhood to restore the facial form and function, but the extent to which it eliminates subsequent conventional procedures remains uncertain. See the image below.
Future and Controversies
As with conventional orthognathic surgery, distraction osteogenesis of the craniofacial skeleton should be considered one of the many tools in the armamentarium of a surgeon. Compared with orthopedic lengthening and rotations of long bones, craniofacial distraction is highly complex. Challenges include careful placement of osteotomies to mobilize facial elements and the multidirectional vectors of distraction. Significant advances in computer-assisted surgery have occurred, and newly designed software can simulate osteotomies and planned distraction in difficult, asymmetrical cases. Cone-beam CT scanners, which provide excellent bony resolution at a fraction of the radiation, are now available for intraoperative CT imaging and postoperative follow-up.
The extent to which distraction osteogenesis will replace conventional approaches depends largely on technical innovations that will allow for implantable multidirectional devices that can be easily activated and controlled remotely with minimal incisions.
Ilizarov GA. The Transosseous Osteosynthesis: Theoretical and Clinical Aspects of the Regeneration and Growth of Tissue. New York: Springer-Verlag. 1992.
McCarthy JG, Schreiber J, Karp N. Lengthening the human mandible by gradual distraction. Plast Reconstr Surg. 1992 Jan. 89(1):1-8; discussion 9-10. [Medline].
Steinberg B, Fattahi T. Distraction osteogenesis in management of pediatric airway: evidence to support its use. J Oral Maxillofac Surg. 2005 Aug. 63(8):1206-8. [Medline].
Denny AD. Distraction osteogenesis in Pierre Robin neonates with airway obstruction. Clin Plast Surg. 2004 Apr. 31(2):221-9. [Medline].
Fritz MA, Sidman JD. Distraction osteogenesis of the mandible. Curr Opin Otolaryngol Head Neck Surg. 2004 Dec. 12(6):513-8. [Medline].
Hong P, Brake MK, Cavanagh JP, Bezuhly M, Magit AE. Feeding and mandibular distraction osteogenesis in children with Pierre Robin sequence: A case series of functional outcomes. Int J Pediatr Otorhinolaryngol. 2012 Jan 12. [Medline].
Li HY, Lee LA. Sleep-disordered Breathing in Children. Chang Gung Med J. 2009 May-Jun. 32(3):247-57. [Medline].
Kaban LB, Seldin EB, Kikinis R, Yeshwant K, Padwa BL, Troulis MJ. Clinical application of curvilinear distraction osteogenesis for correction of mandibular deformities. J Oral Maxillofac Surg. 2009 May. 67(5):996-1008. [Medline].
Polley JW, Figueroa AA. Management of severe maxillary deficiency in childhood and adolescence through distraction osteogenesis with an external, adjustable, rigid distraction device. J Craniofac Surg. 1997 May. 8(3):181-5; discussion 186. [Medline].
Yamauchi K, Takahashi T, Nogami S, Kataoka Y, Miyamoto I, Funaki K. Horizontal alveolar distraction osteogenesis for dental implant: long-term results. Clin Oral Implants Res. 2012 Jan 26. [Medline].
Chin M, Toth BA. Le Fort III advancement with gradual distraction using internal devices. Plast Reconstr Surg. 1997 Sep. 100(4):819-30; discussion 831-2. [Medline].
Nada RM, Sugar AW, Wijdeveld MG, et al. Current practice of distraction osteogenesis for craniofacial anomalies in Europe: A web based survey. J Craniomaxillofac Surg. 2009 May 15. [Medline].
Verlinden CR, van de Vijfeijken SE, Tuinzing DB, et al. Complications of mandibular distraction osteogenesis for developmental deformities: a systematic review of the literature. Int J Oral Maxillofac Surg. 2015 Jan. 44(1):44-49. [Medline].
Qu M, Hou Y, Xu Y, et al. Precise positioning of an intraoral distractor using augmented reality in patients with hemifacial microsomia. J Craniomaxillofac Surg. 2015 Jan. 43(1):106-12. [Medline].
Cohen SR, Boydston W, Burstein FD. Monobloc distraction osteogenesis during infancy: report of a case and presentation of a new device. Plast Reconstr Surg. 1998 Jun. 101(7):1919-24. [Medline].
Figueroa AA, Polley JW, Friede H, Ko EW. Long-term skeletal stability after maxillary advancement with distraction osteogenesis using a rigid external distraction device in cleft maxillary deformities. Plast Reconstr Surg. 2004 Nov. 114(6):1382-92; discussion 1393-4. [Medline].
Grubb J, Smith T. Practical applications of distraction osteogenesis. Am J Orthod Dentofacial Orthop. 2004 Sep. 126(3):271-2. [Medline].
McCarthy JG. Distraction of the Craniofacial Skeleton. New York: Springer-Verlag; 1999.
Menezes RD, Zhao L, Patel PK, Modi V. Volumetric changes in the oropharyngeal airway following bilateral mandibular distraction osteogenesis in Pierre Robin Sequence. J Craniofac Surg. Article submitted July 2008.
Mikhail L, Samchukov JB, Cope A. Craniofacial Distraction Osteogenesis. CV Mosby; 2001.
Molina F, Ortiz Monasterio F. Mandibular elongation and remodeling by distraction: a farewell to major osteotomies. Plast Reconstr Surg. 1995 Sep. 96(4):825-40; discussion 841-2. [Medline].
Wan DC, Nacamuli RP, Longaker MT. Craniofacial bone tissue engineering. Dent Clin North Am. 2006 Apr. 50(2):175-90, vii. [Medline].