Transcutaneous Cardiac Pacing Technique
- Author: Ali A Sovari, MD, FACP; Chief Editor: Richard A Lange, MD, MBA more...
The video below demonstrates transcutaneous cardiac pacing using a defibrillator.
In nonemergent situations, sedate the patient. Record the baseline rhythm and vital signs, if this has not already been done.
Prepare and apply the pads as described earlier (see Positioning). Although transcutaneous pacing has been used continuously for as long as 4-5 days,[14, 15] the sites at which the pads are applied should be changed every 4-5 hours to reduce skin burn and discomfort. For patients who require longer-term pacing, a more appropriate mode of pacing should be considered.
Turn the pacemaker on, and choose the pacing mode. Most pacing units are capable of pacing on either a demand mode or a fixed mode. The demand mode is usually preferable and should be used initially. If capture cannot be obtained on that mode, then the fixed mode should be tried. In the demand mode, the pacemaker senses the intrinsic impulses and delivers current only if necessary.
Capture the heart rate. If the patient is in cardiac arrest with bradycardia or an asystolic rhythm, the pacing should be initiated at the maximum current output to ensure that capture is achieved as soon as possible, after which time the current can be gradually reduced to 5-10 mA above the threshold.
The presence of 1 QRS complex after each pacing stimulus suggests but does not confirm the capture. Cardiac capture should be confirmed by detecting the pulse (see the image below). However, pulse detection can be difficult, particularly in bradyasystolic arrest; in these cases, detecting ventricular capture by means of 2-dimensional echocardiography is a good alternative if the echocardiography system is available.[16, 17]
A small polarization artifact that is usually noted after each pacing artifact should not be confused with a QRS complex. In a healthy individual, the pacing threshold is usually less than 80 mA.
Many factors (eg, obesity, myocardial ischemia, metabolic derangement, pneumothorax, poor skin-to-electrode contact) can increase the pacing threshold, and some of those conditions are correctable. Reevaluate the patient.
It has been shown that walkie-talkies and cellular digital phones can cause various disturbances in the function of external pacing units, including pacing inhibition and asynchronous pacing, even when these devices are as far as 200 cm from the pacing units. Therefore, an appropriate warning against using such communication equipment near patients undergoing transcutaneous cardiac pacing should be issued.
Lee KL, Lau CP, Tse HF, et al. First human demonstration of cardiac stimulation with transcutaneous ultrasound energy delivery: implications for wireless pacing with implantable devices. J Am Coll Cardiol. 2007 Aug 28. 50(9):877-83. [Medline].
Trigano JA, Remond JM, Mourot F, Birkui P, Levy S. Left ventricular pressure measurement during noninvasive transcutaneous cardiac pacing. Pacing Clin Electrophysiol. 1989 Nov. 12(11):1717-9. [Medline].
Feldman MD, Zoll PM, Aroesty JM, Gervino EV, Pasternak RC, McKay RG. Hemodynamic responses to noninvasive external cardiac pacing. Am J Med. 1988 Mar. 84(3 Pt 1):395-400. [Medline].
Murdock DK, Moran JF, Speranza D, Loeb HS, Scanlon PJ. Augmentation of cardiac output by external cardiac pacing: pacemaker-induced CPR. Pacing Clin Electrophysiol. 1986 Jan. 9(1 Pt 1):127-9. [Medline].
Epstein AE, DiMarco JP, Ellenbogen KA, et al. ACC/AHA/HRS 2008 Guidelines for Device-Based Therapy of Cardiac Rhythm Abnormalities: a report of the American College of Cardiology/American Heart Association Task Force on Practice Guidelines (Writing Committee to Revise the ACC/AHA/NASPE 2002 Guideline Update for Implantation of Cardiac Pacemakers and Antiarrhythmia Devices): developed in collaboration with the American Association for Thoracic Surgery and Society of Thoracic Surgeons. Circulation. 2008 May 27. 117(21):e350-408. [Medline].
Ho JD, Heegaard WG, Brunette DD. Successful transcutaneous pacing in 2 severely hypothermic patients. Ann Emerg Med. 2007 May. 49(5):678-81. [Medline].
Sherbino J, Verbeek PR, MacDonald RD, Sawadsky BV, McDonald AC, Morrison LJ. Prehospital transcutaneous cardiac pacing for symptomatic bradycardia or bradyasystolic cardiac arrest: a systematic review. Resuscitation. 2006 Aug. 70(2):193-200. [Medline].
Schwartz B, Vermeulen MJ, Idestrup C, Datta P. Clinical variables associated with mortality in out-of-hospital patients with hemodynamically significant bradycardia. Acad Emerg Med. 2004 Jun. 11(6):656-61. [Medline].
Rosenthal E, Thomas N, Quinn E, Chamberlain D, Vincent R. Transcutaneous pacing for cardiac emergencies. Pacing Clin Electrophysiol. 1988 Dec. 11(12):2160-7. [Medline].
Im SH, Han MH, Kim SH, Kwon BJ. Transcutaneous temporary cardiac pacing in carotid stenting: noninvasive prevention of angioplasty-induced bradycardia and hypotension. J Endovasc Ther. 2008 Feb. 15(1):110-6. [Medline].
Castle N, Porter C, Thompson B. Acute myocardial infarction complicated by ventricular standstill terminated by thrombolysis and transcutaneous pacing. Resuscitation. 2007 Sep. 74(3):559-62. [Medline].
Sodeck GH, Domanovits H, Meron G, et al. Compromising bradycardia: management in the emergency department. Resuscitation. 2007 Apr. 73(1):96-102. [Medline].
Jaworska K, Prochaczek F, Galecka J. Influence of the shape of the pacing pulse on ventricular excitation threshold and the function of skeletal muscles in the operating field during non-invasive transcutaneous cardiac pacing under general anaesthesia. Cardiol J. 2007. 14(2):137-42. [Medline].
Zoll PM, Zoll RH, Falk RH, Clinton JE, Eitel DR, Antman EM. External noninvasive temporary cardiac pacing: clinical trials. Circulation. 1985 May. 71(5):937-44. [Medline].
Falk RH, Zoll PM, Zoll RH. Safety and efficacy of noninvasive cardiac pacing. A preliminary report. N Engl J Med. 1983 Nov 10. 309(19):1166-8. [Medline].
Holger JS, Minnigan HJ, Lamon RP, Gornick CC. The utility of ultrasound to determine ventricular capture in external cardiac pacing. Am J Emerg Med. 2001 Mar. 19(2):134-6. [Medline].
Ettin D, Cook T. Using ultrasound to determine external pacer capture. J Emerg Med. 1999 Nov-Dec. 17(6):1007-9. [Medline].
Trigano AJ, Azoulay A, Rochdi M, Campillo A. Electromagnetic interference of external pacemakers by walkie-talkies and digital cellular phones: experimental study. Pacing Clin Electrophysiol. 1999 Apr. 22(4 Pt 1):588-93. [Medline].
Thomas SP, Thakkar J, Kovoor P, Thiagalingam A, Ross DL. Sedation for electrophysiological procedures. Pacing Clin Electrophysiol. 2014 Jun. 37(6):781-90. [Medline].
Pecha S, Aydin MA, Yildirim Y, et al. Transcutaneous lead implantation connected to an externalized pacemaker in patients with implantable cardiac defibrillator/pacemaker infection and pacemaker dependency. Europace. 2013 Aug. 15(8):1205-9. [Medline].